##### Events

:

:

Elektronik | Funk | Software

Der Technik-Blog

Werbung:

### New Posts

• #### TTGO/LILYGO GPS-Problem

04.07.2022

• #### Ubidots TTN Integration

30.01.2022

• #### Helium Ubidots Integration

26.12.2021

• 26.12.2021

### Events

• Keine zukünftigen Events vorhanden

### The Tech-Blog ### Arduino Power Meter

Alex @ AEQ-WEB
• 13.01.2019

There is also a video on YouTube for this article: Arduino Current Power Sensor - Measure current and power in an AC circuit

In this article, we test about a very cheap inductive AC power meter. Unfortunately, there is no information from the manufacturer how the component works or how accurate the measurement is. In this article we write about our test and the accuracy of this sensor. In addition, we show how the sensor works with an Arduino.

#### How the sensor works

The electrical structure of these inductive power meters is actually very simple and has been used by energy suppliers for decades, above all in the area of high voltage. A big advantage is that with this method high currents can be measured "without contact". The conductor or phase in the AC power grid is passed through this component. When a consumer is connected, a current flows through the line which builds up a magnetic field around the line. Due to this magnetic field, a small voltage is induced in a coil which is located in this component. When more current flows, the induced voltage in the coil is higher. A microcontroller can read the analog voltage, and can convert the input voltage into the flowing current. If you multiply current with the voltage, the power can be calculated in watts.

#### Measure current with an Arduino

For the calculation of power consumption we use an Arduino Uno board. The whole project can also be realized very easily with an ESP32 or NodeMCU. For the calculation, only one analog input on the microcontroller is needed. The input voltage value is then be converted into current and power. In addition, the sensor is still connected to 5 volts and to the ground of the microcontroller.

Werbung:

#### Problems with measuring

A big problem is the correct measurement at AC voltages. The AC grid in Europe has 50 Hz, which means that there are 50 oscillations per second on the measured line. If the analogue input is read out quickly, then you can see that the voltage varies a lot, but very evenly. The following graphic from the Serial Plotter shows the oscillations of the AC power grid. In order to obtain a relatively constant value, the measurement has to run over a certain period of time and from this the mean value must be calculated.

#### Software

The software displays the determined current in the serial monitor and also calculates the power in watts. The measurement is made 2000 times in succession, which then calculates a mean value. This value barely fluctuated in our attempt. Subsequently, the analog value is multiplied by 7. This ensures that you have a real measure. An example: At the analogue input, we get the value 60. At present, 4.2 amperes of current are measured at the load. The output in the Serial Monitor should be 4.20 amps. With a formula ((60 * 7)/100) this value is reached. Thus you get the fixed current value as float-integer.

Werbung:

#### Calibrate the sensor

For the sensor to work properly, it must be balanced with a current meter or a power meter. The voltage at the analog input is not always the same, which is why the formula in the code must necessarily be compared with a reference device.

#### Quality of the sensor

The sensor works very reliable, but is not linear! This means that the sensor delivers an exact result only within a certain range. In our experiment, the sensor had a maximum deviation of 12 watts between 200 and 1500 watts. At under 200 watts or over 1500 watts, the tolerance was much greater. For a consumer consuming exactly 50 watts, the sensor used a consumption of 80 watts, which is clearly above any tolerance limit. It is basically possible to work around this problem with mathematical formulas, but there is a lot of effort behind it to properly calibrate the sensor. In summary, we believe that the sensor works quite well for small projects, but is not suitable for accurate metering or energy billing.

#### Measure three-phase current

With the sensor it is also possible to measure a three-phase network. This requires three sensors, each of which passes through one phase. The individual partial flows must then be added together to determine the total power consumption. On the Arduino board, the sensors are connected to the analog pins from A0 to A2.

###### Current Sensor Code Alex, the founder of AEQ-WEB. He works for more of 10 years with different computers, microcontroller and semiconductors. In addition to hardware projects, he also develops websites, apps and software for computers.

### Top articles in this category: ### PT1000 Converter for Arduino

• Video
• DE/EN

PT1000 sensors can not be measured directly analog with a microcontroller. How to build a PT100 Converter with LM358 and Arduino? ### Arduino LCD Interface

• DE/EN

On this page we show you how to create an interface for a TFT display with an Arduino. You can download source code and library here

Werbung:

### New Posts

• #### TTGO/LILYGO GPS-Problem

04.07.2022

• #### Ubidots TTN Integration

30.01.2022

• #### Helium Ubidots Integration

26.12.2021

• 26.12.2021

### Events

• Keine zukünftigen Events vorhanden